Installation of the surgical shadowless lamp Debugging inspection: In this process, the main thing is to see that all the screws are in place and tightened during the installation process, whether the various decorative covers have been covered, or whether there is any other installation missing.
Circuit inspection of surgical shadowless lamp products: Circuit safety inspection is a key point of safety inspection for surgical shadowless lamps: first, check whether the shadowless lamp has a short circuit or open circuit under power-off conditions. If not, check whether the power supply of the shadowless lamp is stable and the transformer is not present. Whether the input voltage is stable and meets the requirements of shadowless lights.
Adjustment of the balance arm of the surgical shadowless lamp: When the medical staff adjusts the position of the surgical shadowless lamp, the balance arm system is required for force. When the medical staff adjust the position of the surgical shadowless lamp, the balance arm system is required to bear the force. Therefore, it is necessary to check whether the balance arm can be adjusted to the angle required by the medical staff and whether it can bear the force.
Debugging and inspection of joint flexibility of surgical shadowless lamp: Because the angle of the shadowless lamp needs to be adjusted, the flexibility of the joint is also very important, mainly to adjust the damping screw of the joint. The standard stipulates that the tightness of the damping adjustment is that the force of pushing or rotating the joint at any position is close to 20N or 5Nm.
Operation shadowless lamp adjusts the depth of illumination brightness: because doctors are likely to need to see the depth of the patient's wound during surgery, the surgical shadowless lamp is required to have a good illumination depth, generally 700-1400mm is better.
Surgical shadowless lamp color temperature check: This is the more important point of surgical shadowless lamp. Good illuminance and color temperature help doctors carefully observe the patient’s wound tissue, distinguish organs, blood, etc., so the illuminance close to sunlight and the color temperature of 4400-4600K are compared suitable